Log in. Shopping cart 0. You have no items in your shopping cart. Manufacturer part number: SD. X Available Configurations. Overview Specifications Documentation Included Accessories. The SD is designed for installing, provisioning, maintaining, and troubleshooting wireless base station cable and antenna systems. Because the Site Master was designed specifically for field environments, it can easily withstand the day-to-day punishment of field use.
The analyzer is almost impervious to the bumps and bangs typically encountered by portable field-equipment. Site Master operation is straightforward; measurements are obtained through a menu-driven user interface that is easy to use and requires little training.
The large, and high-resolution TFT color display makes test interpretation easy and quick. A full range of markers enables the user to make accurate measurements. Data Sheet. It dramatically increases your productivity and transforms the traditional fix-after-failure maintenance model to one that identifies and fixes problems before major failures occur. Sweep Gear. Sweep Tools.
Anritsu SP Site Master. Data analysis software enables assessment of system trends, problems, and performance in addition to professional report generation. FlexCal broadband calibration 25 MHz to 4 GHz allows troubleshooting cable and antenna systems without multiple calibrations and calibration setups.
Built-In Worldwide Signal Standards eliminates the need to remember system frequencies and manually input the start and stop frequency values. Built-In Average Cable Loss eliminates the need to calculate a measured cable loss value. Built-In Power Meter Option measures the base station transmitter power accurately.
Having trouble viewing this file? Click this link to download. Skip to the end of the images gallery. Skip to the beginning of the images gallery. Availability: In stock. SKU 6C Add to Cart. Add to Wish List Add to Compare. Features Handheld, battery-operated design Weighs less than 5 lbs. Additional Information.
Features new to the S33xD models are not available here. To access the new features, use Control Byte 33 21h. This command cannot be used with a frequency converter module attached Option 6 required. DTF windowing allows you to make a trade off between side lobe height and resolution.
Page 59 2. Page 76 3. Also returns power meter settings. If Option 6 is installed and the frequency converter module is attached, the center frequency should be scaled by the scale factor of the module. Each antenna can have up to 60 antenna factors. Send a 0 zero following the command to read the current channel power measurement i. Send to read the channel power associated with a stored trace use Query Trace Names, 24, to obtain trace numbers. If Option 6 is installed and the frequency converter module is attached, the frequencies should be scaled by the scale factor of the module.
Single Sweep Mode activates once the Site Master exits from the remote mode. To access the new features use Control Byte 8Fh. If Option 6 is installed and the frequency converter module is attached, the frequency will be scaled by the scale factor of the module. If it is, it then checks to see if it is at the correct frequency for PLL locking. Bytes to Follow: 1 byte Attenuation 0 — An invalid setting returns the baud rate to The resolution is 0. Use this command for both Spectrum Analyzer and Power Meter modes.
These include functions to set up the comm port, enter and exit remote mode, and set the reference level of the spectrum analyzer. This manual is also suitable for: Site master sd. Print page 1 Print document pages.
Rename the bookmark. Delete bookmark? Cancel Delete. Delete from my manuals? Sign In OR. Don't have an account? Sign up! Restore password. Upload manual. Table of Contents Programming Overview. Upload Sweep Trace — Control Byte 36 24h For information on firmware upgrades, contact your local Anritsu Service Center.
Commands listed in this manual are not all backward-compatible with earlier Anritsu models. Remote mode differs from normal repetitive sweep and single-sweep modes. During remote mode, the. The Site Master sends a response byte of FFh then exits remote mode. Remote Mode Changes to Site Master Operating Parameters System parameters changed during remote mode remain changed for normal operation after the unit exits remote mode.
However, the changes are not automatically written to the non-volatile. The current value of the flags can be obtained by executing command 29, Query System Setup, and parsing the values from the appropriate bytes. The Site Master acts on the entire byte. So, the state of each of the bits must be defined every time the command is issued. The response byte will not be sent until the mode change is complete. See control byte 29 1Dh response byte 3 for the current Site Master measurement mode. See Control Byte to set markers in Spectrum Analyzer mode.
Site Master Handheld Cable and Antenna Analyzer Model SD covers 25 MHz to MHz spectrum. Suited for field technicians of any experience level working. Anritsu SiteMaster SD Cable & Antenna Analyzer (Handheld) 25MHz - 4GHz. Site Master™ Handheld Cable and Antenna Analyzer Model SD covering the User Guide. Site Masterä. SD. Cable and Antenna Analyzer. Anritsu Company. PN: Jarvis Drive. Revision: L. Morgan Hill, CA